Risk of electric shock due to false readings from product
Fluke state that:
“It has been determined that certain Recalled Testers may experience premature failure of the interconnecting cable under cyclical bending stress. Recent experiments conducted by Fluke have shown a wide distribution in the number of cycles to failure exhibited by the Recalled Testers, meaning that certain Recalled Testers are subject to failure earlier than expected, leading to a useful life for certain of them that Fluke considers unacceptable.
Further, the cable may fail in an intermittent fashion where, depending on the cable flex orientation, it is possible for a Recalled Tester to pass a self check continuity test or validation on a known voltage source, while subsequent tests may display a false negative. The primary function of a T-Pole Tester is to detect the presence or absence of voltage to determine whether it is safe to touch an electrical installation and begin work.
A false negative may lead to electric shock or arc flash from subsequent user actions as a result of the false negative indication, which may lead to injury or even death. Because of this risk, please IMMEDIATELY STOP using your T-Pole Tester”
What To Do
If you believe that you possess an affected product stop using it immediately and check the below webpage for further details and recall information